Agilent technologies HP B Pdf User Manuals. View online or download Agilent technologies HP B Operating Manual. HP B Synthesizer Function Generator. Service Manual. Up for sale is the following Technical Manual in original binder/binding Original Manufacturer’s. I recently won a “for parts or repair” HP B function generator for a great Although free scans of the B service manual are readily.
|Published (Last):||22 July 2005|
|PDF File Size:||19.38 Mb|
|ePub File Size:||9.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
The gate output drives a serbice amplifier which inverts the signal and the resulting current pulse signal is sent through a filter which shapes the triangle. If amplitude problems are encountered, it is important to disable this auto calibration.
A method for obtaining this frequency is illustrated in Figure Measure the voltage at the junction of R41 and R Function Circuits Service Group J. Check the machine data bus lines at the input manuxl output of A15U for logic level transitions.
The signal level from the output amplifier can be tested in the level comparator to determine if a level correction is needed, thus providing an automatic amplitude calibration.
Let us assume that this is true, and that the -h N number is List of Manufacturers 332b5 for burned or loose components. Adapter Cable An over-voltage protection circuit in the unregulated -i-5V supply prevents the voltage from becoming high enough to damage the supply circuits. This cable has a gp plug at one end to connect to the phono jacks used as signal coimectors on the printed circuit board.
After blinking the LEDs, the test loops repeatedly fortroubleshooting. A14 Assembly Signatures 8-J-5 8-N-l. IV, 10V, V Accuracy: Please enter a valid ZIP Code.
Learn More – opens in a new window or mahual Any international shipping is paid in part to Pitney Bowes Inc. Attenuation and Voltage Ranges If not, A21U19 is probably defective.
Test Equipment Required for Adjustments Program the front panel for a sine function at lOVp-p. Set the amplitude to 3.
For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
T-9 ‘ Indicates factory selected values See introduction to this section for ordering information Table 3325b Mouse over to Zoom – Click to enlarge.
To check for stuck keys, press Shift, Deg, Self Test, 2, 0. At turn-on, maual waveform should be a 2Vp-p, 30MHz sine wave on both leads. Will usually ship within 2 business days of receiving cleared payment – opens in a new window or tab. Also, when replacing the shield nearest the front of the instrument, make sure the hole in the cover is over the mixer adjustment resistor.
Shift DegmVrms Self Test 5 3 b. When the current source turns off, the integrator voltage is sam- pled and held. Visit my eBay store. The resistance should range from to Set frequency to 21 MHz.